Humidity and hygrometers
It's that time of year again. This post got me curious:
Here's a link to a nice set of instructions for doing the "salt test." You can do the test with just about any hygrometer and note the error. If you want a hygrometer that you can calibrate, here's a nice little one.
Some of us obsess about maintaining proper humidity. The hygrometers most of us end up with can be pretty far off.
Thanks for the link on the hygrometer. I've used the wet/dry bulb technique, but it's a hassle to set up, and the electronic version seems convenient.
I reproduced the instructions below from this post. The original site can take a bit of time to load. Kudos to Tman at puff.com.
How to do a salt test
This is how I do my salt test. Salt has an amazing property where RH% will be pretty much nearly be 75% when it is in a saturated condition with water.
(Look for Sodium Chloride, aka. Salt)
Here's what you'll need:
Salt (I use Morton Salt)
Small Ziplocİ Bag (Small as possible)
Bottle cap (Gatorade cap works best. Wide&shallow works best. I'll use water bottle cap for this)
Toothpick (whatever to mix the salt)
Hygrometer to be calibrated
First, put salt in the bottle cap
Then, put couple drops of water in the salt. Notice it's partially wet. You want the salt to suck up all the water.
Mix the salt/water mix so it will be a good mix.
Once you have the saturated salt, put it into a Ziploc bag with your hygrometer. Pinch up the center of the Ziploc bag so it will inflate the bag. While you are doing this, seal the bag with your other hand. The bags should be inflated so it won't cover the sensor opening of the hygrometer or the bottle cap.
Give it couple of hours for the RH% to stabilize. I think 6 hours should be sufficient time to do this.
Once the RH% stabilizes, calibrate your hygrometer within the Ziploc bag. On HygroSet, I'll have to twist the dial (clockwise for +RH% and counter-clockwise for -RH%. Press the Set button.)
On Xikar hygrometer, press "Calibrate".
It might be easier to use a tip of a pen to press the button. If you want, you can leave the bag for another 6 hours and make sure the reading stabilizes at 75%.
That's it! Enjoy your calibrated hygrometer!
Amazing, somebody is selling a "one step hygrometer calibration kit" on Amazon. And guess what... it maintains a constant humidity of 75 percent.
Nice to see a quote of one of my postings starting a new thread :D
This is the YouTube video I used:
It is essentially the same but now on video. And with a plastic box instead of a bag.
Anyway... thanks again. :)
Fascinating that there is such a simple solution(pun intended) to hygrometer calibration. I would point out however that just because your hygrometer is accurate at 75% doesn't mean the same is true at 40%. I wonder if there's a way to test accuracy at lower humidity. I'm lucky that where I live humidity is a non-issue. My apartment never dips below 40%, and the handlers in my former orchestra leave basses and cellos on the truck overnight in sub-freezing weather all the time. As unpleasant as it is to warm up a chilled bass before a morning rehearsal, temperature cracks and open seams occur only very rarely.
|All times are GMT -6. The time now is 04:30 AM.|
Powered by vBulletin® Version 3.6.12
Copyright ©2000 - 2014, Jelsoft Enterprises Ltd.